As promised I would let all those who were interested Omron UK's response in respect of the intermittent stop and starting with my Omron MicroAir. Tests have shown that whilst the Mesh Cap was in good working order, that the contacts under the medication unit had some corrosion, which was causing the problem with the flow etc. This could have been dealt with by servicing I have since been informed, but since this last MicroAir was just 4 months old and the problem had been occurring for much of that time, I still feel disappointed after having previously owning an excellent long serving MicroAir.
I have explained to Omron today via telephone call about my previous post on this forum and how some others have had difficulties too. Omron are a little disappointed that I have experienced these difficulties and have informed me that should any of those you feel you need to discuss your own issues with the Omron MicroAir please get in touch.
My Point of Contact Was:
Kate Panter, Office Manager, Omron UK, Telephone Number 0870 7502771.